The Analysis Packages for Microscope Digitalization allows you to retrofit your existing laboratory microscope with an automatic cell counting functionality.
A laboratory microscope with a 40X objective
A Neubauer / Thoma chamber for cell counting
The compatible camera which can be bought directly from the manufacturer or alternatively from Oculyze
- An active Oculyze account
A cable to connect the compatible mobile phone with the camera
If you have a device with USB-C: USB-C (male) to USB-B (male) cable OR USB-A (female) to USB-C (male) adapter
If you have a device with Micro USB: USB-A (female) to Micro-USB (male) adapter
Setting up the External Camera
The first step to connect the camera to the microscope. If your Microscope supports it, please connect the camera can to the C-Mount, otherwise you can connect it using the eye piece. If you are unsure if your microscope has a C-Mount port, please check the manufacturer specifications.
After mounting the camera it can be connected to the android device.
The videos below show both the process for both types of microscopes
Calibrate for your setup
After the Camera has been mounted it needs to be to be calibrated. The process is shown in the below
Once the External Camera option is enabled, the device is ready for calibration. This operation is needed only the first time the device is used with your microscope, and has to be repeated only when using another microscope or when changing magnification.
After connecting the camera correctly, position the counting chamber on the microscope plate, focus on the counting grid. Once the grid is in focus, move it to find its center, this will help the correct calibration. When the squares of the grid are in focus, press the capture button and wait until the correct square is detected and marked by the white overlaying square. Press “Keep” only if the correct square (the one corresponding to the size shown in the chamber data). When a different rectangle or none is detected, repeat the process, refocusing and cleaning the chamber if needed.
Once the detection is successful, you’ll be prompted to input the chamber height and square size. Insert this data and press OK to finalize the calibration.
The sample preparation and chamber loading does not differ from your current manual procedure, if you are unsure about these steps you can review the corresponding section in our manual for the BB2 microscope.
Chamber preparation, loading and cleaning
For instructions on how to prepare, load and maintain you counting chamber please review the manufacturer instructions.
To measure the sample please follow the instructions as described in the manual for the BB2 App.
The grid of the counting chamber is not needed after the initial calibration. The Oculyze Image recognition is designed to ignore the grid if present and usually it will not interfere with your measuarement, for even better results however we recommend taking the images in an area of the chamber without the grid.
Note that it is essential that you have set the App to external Camera mode and calibrated it to your setup as shown in the video above.
For additional support please write to: firstname.lastname@example.org
Please note that using Oculyze system with external camera has limited support compared to the standard Oculyze Better Brewing 2.0 microscope kit. Customer is responsible about calibration according to these instructions.